Instruction/ maintenance manual of the product 2100 Furuno
Go to page of 10
1 ECDIS FEA-2100/2105 C-M AP Chart Update M anual ・ Up date char t b efor e em bar ki ng on a v oyage. D o not up date char t s duri ng a v oyage. ・ Dur ing updat ing , do not disp lay the r adar ov er lay scr een or the r oute m oni tor sc reen.
2 3. C heck the b ox on [Autom atic SENC conv ersi on] and [Autom atic D isp lay Uni t]. (Or c onfi rm that the chec k m ark s are di sp layed.) T o check t he box , pl ace the c ursor on the b ox and pr ess t he [SELECT] key . 4. Pl ace the cur sor on t he [Cl ose] b utton and pr ess the [SELEC T] key to close the w i ndow s.
3 The catalogue nam e [CM 93PROF 3] is show n. 5. Plac e the curs or on the [O K] but ton and pr ess the [SELECT] key . Note: If the catalog ue nam e is t he sam e as an exi sting one you nam ed pr ev iousl y , the f ollow i ng m essag e ap pear s.
4 3. Display ing th e Catalogu e Char t 1. Pres s the [CH ART PLAN ] key to disp lay the m enu. 2. Plac e the curs or on the [C atalogue] and pr ess the [SELECT] key . 3. Chec k on the [N am e] and [Catalog ue char t s] check box es and sel ect [C D] in t he [View fi lter s] w indow.
5 6. Set the c hart g roup to up date as f oll ow s. a. Plac e the curs or on the [G roups] button at the b ottom of the [Catalog ue] w i ndow and press the [S ELECT] ke y . b. Place the cur sor on the [New ] b utton and press the [S ELECT] ke y .
6 e. Plac e the curs or on the [A dd Fram e] button and pr ess t he [SELECT] key . f . Rotate the tr ackb all to pl ace the c ursor ins ide the char t cel l f ram e des ir ed and pr ess the [SELECT] key . R epeat this to select all desi red ar eas.
7 4. Char t Conv ersion (fr om E NC for mat to SE NC for mat) The c hart d ata in a C D- ROM (ENC f orm at) mus t be conv er ted to SENC form at i n order to disp lay it on t he ECDI S. Note that t he conv ersi on takes sev er al hours .
8 6. Plac e the curs or on the [M is sing only] chec k b ox and pr ess t he [SELECT] key to dis pl ay the check mark. 7. Plac e the curs or on the [Load] button and p r ess the [SELECT] key . The catalog ue data is l oaded to t he EC DIS.
9 After loading all char t s desi red, t he chart conv ers ion ( fr om ENC to SENC ) autom ati call y st art s. 8. At this tim e, the m essage [No Base Cell av ail abl e] and chart cell num ber s ar e disp layed in the [ S57 LOAD] w i ndow.
10 The err or m ess a ge m ay app ear dur ing conv ers ion, how ever leav e the m ess age di splayed until the conv er sion i s com p leted. The list i n the [Conv ersion List] b ecomes blank after conv er sion i s com p leted.
An important point after buying a device Furuno 2100 (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought Furuno 2100 yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data Furuno 2100 - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, Furuno 2100 you will learn all the available features of the product, as well as information on its operation. The information that you get Furuno 2100 will certainly help you make a decision on the purchase.
If you already are a holder of Furuno 2100, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime Furuno 2100.
However, one of the most important roles played by the user manual is to help in solving problems with Furuno 2100. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device Furuno 2100 along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center